Shenzhen ThreeNH Technology Co., Ltd.
Leader Of Color Management In China
|Place of Origin:||P.R.China|
|Certification:||CE / ROHS / FCC / ISO9001:2015,TUV|
|Minimum Order Quantity:||1pcs|
|Packaging Details:||Packed in hard nylon box firstly, and then reinforced with carton box for outer packing|
|Delivery Time:||1 working day|
|Payment Terms:||T/T, Paypal, Western Union|
|Supply Ability:||5000pcs per month|
|Measure Angle:||60°||Measuring Area (mm):||9X15|
|Standard:||Comply With Standard ISO 2813、GB/T 9754、ASTM D 523、ASTM D 2457||Errors:||±1.2 , ± 1.2%|
|Language:||Chinese/English||Software:||GQC6 Quality Control Software With QC Report Printing Functions And More Extended Functions.|
Digital Touch Screen Gloss Measurement Instruments NHG60 60 Degree For Ink Resin Skin Texture
NHG60 Gloss Measurement Instruments
NHG60 Gloss Measurement Instruments features with 60° angle, manufactured according to ISO2813 and GB/T 9754. It is also compatible with the standards of ASTM D523, ASTM D2457. NHG60 conforms to JJG696 first class gloss meter working requirement, corresponding with CIE 1931(2°) under CIE C light source.
NHG60 gloss meter can test material with gloss (0-1000Gu), and universally apply to paint, ink, stoving varnish, coating, wood products; marble, granite, vitrified polished tile, pottery brick and porcelain; plastic, paper; hardware industries, etc. for gloss measurement and gloss data transmission.
1. 60° angle for semi gloss
2. Display 5 sets of measurement data, good for comparison
3. Measuring mode: basic measurement, statistical measurement, continuous measurement, QC measurement (only for GQC6 software).
4. Continuous mode to check uniformity over large areas
5. Connect to PC, more extend functions
6. Input gloss value manually
7. Large touch screen operation, easy to use
8. Beauty appearance, good man-machine communication interface
9. Powerful functions to meet many different requirements
10. Large storage to save over 5000 data
11. Built-in lithium ion rechargeable battery.
NHG60 Gloss Measurement Instruments can be used in :
paint, ink, stoving varnish, coating, wood products; marble, granite, vitrified polished tile, pottery brick and porcelain; plastic, paper; hardware industries, etc. for gloss measurement and gloss data transmission.
|NHG60 60° Precise Gloss Meter|
|Comply with standard ISO 2813,GB/T 9754,ASTM D 523,ASTM D 2457|
|Measuring Area (mm)||9X15|
|Measurement Range Repeatability Reproducibility||0-10GU 10-100GU 100-1000GU|
|± 0.1GU ± 0.2GU ± 0.2%GU|
|± 0.2GU ± 0.5GU ± 0.5%GU|
|Measurement Standard||Conform with JJG696 first class gloss meter working requirement|
|Chromaticity Corresponding||Corresponding with CIE 1931(2°) under CIE C light source|
|Errors||±1.2 , ± 1.2%|
|Dimension||L*W*H : 160mm*75mm*90mm|
|Battery||3200mAh Li-ion Battery, >10000 times(within 8 hours)|
|Viewing Screen||TFT 3.5 inch, capacitive touch screen|
|Data Save||Basic mode: 1000, Statistical Mode: 5000, Continuous mode : 5000|
|Software||GQC6 Quality Control Software with QC report printing functions and more extended functions.|
|Humidity||<85% relative humidity, no condensation|
|Standard Accessories||Power Adapter, USB cable, User Manual, CD (Including QC software), Calibration Plate|
|Optional Accessories||Miniature Printer|
|Note||The specifications are subject to change without notice.|
How to choose gloss meter angle ?
How to choose gloss meter angle ? For early contact with the gloss meter reader is a more confused proposition, it comes out a lot when searching on the Internet. There is single-angle gloss meter, three-angle gloss meter, a 45-degree gloss meter, a 60-degree gloss meter, so which gloss meter is suitable for us to use it?
In general, gloss meter has 20 °, 45 °, 60 °, 75 °and 85 °specifications. Then how to choose gloss meter angles ? If our customers required us what angle gloss meter to measure the gloss data then we need choose according to the customer's request.
If the customer didn’t specify what angle, it can choose 60 °general version gloss meter, 20 °angle meter is generally used in the gloss products higher than 70Gu, 85 °angle is generally used in gloss products less than 10 °, 45 °And 75 °angles used less, 45 °gloss meter is mainly used for the surface gloss measurement of plastic film, 75 °gloss meter is mainly used for enamel surface gloss measurement.
|Gloss Range||60° Value||Notes|
|High Gloss||>70 GU||If measurement exceeds 70 GU, change test setup to 20°|
|Medium Gloss||10 - 70 GU||If measurement during 10- 70 GU, change test setup to 60°|
|Low Gloss||<10 GU||If measurement is less than 10 GU, change test setup to 85°|
Shenzhen ThreeNH Technology Co., Ltd
Add: 4/F, Building 8, Nangang Second Industry Zone, Xili, Nanshan District, Shenzhen, 518055 China.
Tel: 0086-755-27198826 Ext: 601