Shenzhen ThreeNH Technology Co., Ltd.
Leader Of Color Management In China
|Place of Origin:||P.R.China|
|Certification:||CE / ROHS / FCC / ISO9001:2015,TUV|
|Minimum Order Quantity:||1pcs|
|Packaging Details:||Packed in hard nylon box firstly, and then reinforced with carton box for outer packing|
|Delivery Time:||1 working day|
|Payment Terms:||T/T, Paypal, Western Union|
|Supply Ability:||5000pcs per month|
|Dimension:||L*W*H : 160mm*75mm*90mm||Weight:||350g|
|Language:||Chinese/English||Software:||GQC6 Quality Control Software With QC Report Printing Functions And More Extended Functions.|
|Standard Accessories:||Power Adapter, USB Cable, User Manual, CD (Including QC Software), Calibration Plate||Measuring Range:||20°:0~2000GU 60°:0~1000GU 85°:0~160GU|
Handheld Gloss Measurement Instruments NHG268 Multi Angle Gloss Tester for Furniture Paint Ceramic
The reason for the influence of the gloss of the marble is the size of the crystal grains. The larger the grain size, the lower the gloss. The 3nh HG60 marble gloss meter can be used to detect the gloss of the marble surface.
From the perspective of the mineral structure of the marble, the grain size affects the gloss of the marble. The larger the crystal grain size, the lower the gloss obtained and the worse the polishing performance. The more uniform the crystal particles, the better the polishing performance of the marble. There are also differences in the gloss of the polished surface of the marble, that is, the gloss has directionality. The polished surface of the medium and fine grained stone has good consistency in different directions, and the gloss of the polished surface of the marble is improved. The difference in different directions is decreasing.
The gloss of the marble will become dull with time, so we need to maintain the marble regularly to maintain its gloss. The 3nh HG268 marble gloss meter can be used to check the gloss of the marble surface.
Marble is very popular because people likes its gloss and natural texture, but the gloss of the marble will become dull as time goes by, how can we maintain the marble?
1. Avoid the marble in the oil, because the marble is easy to absorb oil and spills.
2. Maintain basic protective agent when using marble.
3. Regularly clean the marble (use a mild product to clean it, because acidic cleaners will corrode the surface of the marble to reduce its gloss)
4. If the marble has been damaged, use a steel ball to wipe the previous gloss after polishing with an electric sander.
5. Regular waxing to make the gloss more durable
Handheld Gloss Measurement Instruments NHG268 gloss meter features with 20, 60 and 85° angle, manufactured according to ISO2813 and GB/T 9754. It is also compatible with the standards of ASTM D523, ASTM D2457. NHG268 conforms to JJG696 first class gloss meter working requirement, corresponding with CIE 1931(2°) under CIE C light source.
NHG268 gloss meter can test material with high gloss (0-2000Gu), and universally apply to paint, ink, stoving varnish, coating, wood products; marble, granite, vitrified polished tile, pottery brick and porcelain; plastic, paper; hardware industries, etc. for gloss measurement and gloss data transmission.
1. 20, 60 and 85° in one: from high to low gloss - have the specified angle on hand
2. One button for all angles measurement at the same location
3. Display 5 sets of measurement data, good for comparison
4. Measuring mode: basic measurement, statistical measurement, continuous measurement, QC measurement (only for GQC6 software).
5. Continuous mode to check uniformity over large areas
6. Connect to PC, more extend functions
7. Input gloss value manually
8. Large touch screen operation, easy to use
9. Beauty appearance, good man-machine communication interface
10. Powerful functions to meet many different requirements
11. Large storage to save over 5000 data
12. Built-in lithium ion rechargeable battery.
|Comply with standard ISO 2813, GB/T 9754, ASTM D 523, ASTM D 2457|
|Measuring Area (mm)||20°: 10x10 60°: 9x15 85°: 5x36|
|Measuring Range||20°:0~2000GU 60°:0~1000GU 85°:0~160GU|
|Measurement Range||0-10GU 10-100GU 100-2000GU|
|Repeatability||± 0.1GU ± 0.2GU ± 0.2%GU|
|Reproducibility||± 0.2GU ± 0.5GU ± 0.5%GU|
|Measurement Standard||Conform with JJG696 first class glossmeter working requirement|
|Chromaticity Corresponding||Corresponding with CIE 1931(2°) under CIE C light source|
|Errors||±1.2 , ± 1.2%|
|Dimension||L*W*H : 160mm*75mm*90mm|
|Battery||3200mAh Li-ion Battery ,>10000 times(within 8 hours)|
|Display||TFT 3.5 inch, capacitive touch screen|
|Storage||Basic mode: 1000; Statistical Mode: 5000,Continuous mode : 5000|
|Software||GQC6 Quality Control Software with QC report printing functions and more extended functions.|
|Humidity||<85% relative humidity, no condensation|
|Standard Accessories||Power Adapter, USB cable, User Manual, CD (Including QC software), Calibration Plate|
|Optional Accessories||Miniature Printer|
|Note||The specifications are subject to change without notice.|
Gloss Measurement Instruments is widely used in Paint Paper ink Plastic film Bamboo flooring Stainless steel Aluminum oxidation Stone Ceramic tile Leatherwear...
Why Measure Gloss ?
Gloss is an aspect of the visual perception of objects that is as important as color when considering the psychological impact of products on a consumer. In other words, "Gloss Sells" Gloss has been defined as the attribute of a surfaces that causes it to have a shiny or lustrous, metallic appearance.
The gloss of a surface can be greatly influenced by a number of factors, for example the smoothness achieved during polishing, the amount and type of coating applied or the quality of the substrate. Manufacturers design their products to have maximum appeal. Such examples are; highly reflective car body panels, glossy magazine covers or satin black designer furniture. Now what happens when products all of a sudden look different? Customers see this as a defect, or poor quality. Using a gloss meter and having good quality control practices eliminates this variable as a problem.
It is important therefore that gloss levels be consistent on every product or across different batches of products. Gloss can also be a measure of the quality of a surface, for instance a drop in the gloss of a coated surface may indicate problems with its cure- leading to other failures such as poor adhesion or lack of protection for the coated surface.
It is for these reasons that many manufacturing industries monitor the gloss of their products, from cars, printing and furniture to food, pharmaceuticals and consumer electronics. We also have Gloss Sensors used in process for real time gloss measurement.
Gloss Meter by Industry
Many industries measure gloss to ensure the quality of their products. It has been shown that the gloss level of a product has as much psychological impact on a consumer as its colour. Many products are designed with combinations of high gloss and matte finishes, maximizing their customer appeal.
Surface quality can often be assessed by measuring gloss. For example poorly cured paints or coatings can exhibit unusual gloss levels and can lead to chipping and flaking at a later stage in manufacture. Monitoring a product's gloss means that problems such as these can be highlighted early, avoiding costs of re-coating or recalling products.
|APPLICATION||Coatings, plastics and related materials||Ceramic,
|High Gloss||Semi Gloss||Low Gloss||Semi Gloss||Low Gloss|
|DIN EN ISO 2813||YES||YES||YES|
|ASTM D 523||YES||YES||YES|
|ASTM D 2457||YES||YES||YES||YES|
|JIS Z 8741||YES||YES||YES||YES||YES|
|ASTM C 346||YES|
|Tappi T 480||YES|
|En ISO 7668||YES||YES||YES||YES|
Shenzhen ThreeNH Technology Co., Ltd
Add: 4/F, Building 8, Nangang Second Industry Zone, Xili, Nanshan District, Shenzhen, 518055 China.
Tel: 0086-755-27198826 Ext: 601