Shenzhen ThreeNH Technology Co., Ltd.
Leader Of Color Management In China
|Place of Origin:||China|
|Minimum Order Quantity:||1pcs|
|Packaging Details:||Carton package|
|Delivery Time:||1 working days|
|Payment Terms:||T/T, Western Union,paypal|
|Sensor:||256 Image Element Double Array CMOS Image Sensor||Light Source Device:||360nm-780nm Combined LED Lamp, 400nm Cut-off, 420nm Cut-off, UV Lamp|
|Wavelength Range:||360-780nm||Measuring Aperture:||Reflective : Φ30mm/Φ25.4mm, Φ18mm/Φ15mm, Φ10mm/Φ8mm, Φ6mm/Φ4mm;|
3nh Lipstick color Experimental data analysis Benchtop grating spectrophotometerYS6010
YS6010 is a benchtop grating spectrophotometer which is developed by 3nh independently with independent intellectual property. YS6010 has many features, like 7 inches TFT capacitive touch screen display, full illuminants, reflective d/8 and transmissive d/0 geometry(including or excluding UV). With very stable and precise color measurement, large storage and powerful PC software all makes YS6010 ideal for color analysis within R&D and laboratory environments.
YS6010 benchtop spectrophotometer is used to do precise color analysis and transmission in laboratories. It can be widely applied in different industries, such as plastics, electronics, paint and ink, printing, garments, leather, paper, auto, medical, cosmetics, food, science institutes, laboratories.
|Model||YS6010 Benchtop Spectrophotometer|
|Optical Geometry||Reflect: di:8°, de:8°(diffused illumination, 8-degree viewing angle) ; SCI (specular component included)/SCE (specular component excluded) ; Include UV / excluded UV light source Transmittance: di:0°, de:0° (diffuse illumination: 0° viewing) ; SCI (specular component included)/SCE (specular component excluded) ; Include UV / excluded UV light source; Conforms to CIE No.15,GB/T 3978,GB 2893,GB/T 18833, ISO7724/1, ASTM E1164, DIN5033 Teil7|
|Application||It is used for accurate analysis and transmission of laboratory color. Apply in paints, inks, textiles, garments, printing and dyeing, printing etc industries for color transfer and quality control.|
|Integrating Sphere Size||Φ154mm|
|Light Source||360 nm to 780 nm, Combined LED Light, 400nm cut-off light source|
|Spectrophotometric Mode||Concave Grating|
|Sensor||256 Image Element Double Array CMOS Image Sensor|
|Measured Reflectance Range||0-200%|
|Measuring Aperture||Reflectance: Φ30 mm/Φ25.4 mm; Φ10 mm/Φ8 mm; Φ6 mm/Φ4 mm|
|Transmittance: Φ30 mm/Φ25.4 mm; Remark: 1. Automatic identification of switch caliber 2. Customized Configuration caliber and lens position|
|Specular Component||Reflectance: SCI&SCE / Transmittance: SCI&SCE|
|Color Space||CIE LAB,XYZ,Yxy,LCh,CIE LUV,Musell,s-RGB,HunterLab,βxy,DIN|
|Color Difference Formula||ΔE * ab ,ΔE * uv ,ΔE * 94 ,ΔE * cmc(2:1) ,ΔE * cmc(1:1) ,ΔE * 00 , DINΔE 99 ,ΔE (Hunter), ΔE * ab ,ΔE * uv ,ΔE * 94 ,ΔE * cmc(2:1) ,ΔE * cmc(1:1) ,ΔE * 00 , DINΔE 99 ,ΔE (Hunter)|
|Other Colorimetric Index||WI(ASTM E313, CIE/ISO, AATCC, Hunter),|
|YI(ASTM D1925, ASTM 313),TI(ASTM E313, CIE/ISO),|
|Metamerism Index MI, Staining Fastness, Color Fastness, Color Strength, Opacity, Cobalt platinum index, Gardner, 8° Glossiness,555 tone classification|
|Displayed Data||Spectrogram/Values, Samples Chromaticity Values, Color Difference Values/Graph, PASS/FAIL Result, Color Offset|
|Measuring Time||About 2.4s (Measure SCI & SCE about 5s)|
|Repeatability||Spectral reflectance: Φ25.4mm/SCI, Standard deviation within 0.05% (400 nm to 700 nm: within 0.04%)|
|Chromaticity value:Φ25.4mm/SCI, Standard deviation within ΔE*ab 0.02 ( When a white calibration plate is measured 30 times at 5 second intervals after white calibration) Spectral Transmittance: Φ25.4mm/SCI, Standard deviation within 0.05% (400 nm to 700 nm: within 0.04%)|
|Spectral transmittance: Φ25.4mm/SCI, Standard deviation within 0.06% (400 nm to 700 nm: within 0.04%)|
|Inter-instrument Error||Φ25.4mm/SCI, Within ΔE*ab 0.15(Average for 12 BCRA Series II color tiles)|
|Measurement Mode||Single Measurement, Average Measurement|
|Locating Method||Camera View Finder Locating|
|Power||AC 24V, 2A Power adapter power supply|
|Illuminant Life Span||5 years, more than 3 million times measurements|
|Display||7-inch TFT color LCD, Capacitive Touch Screen|
|Data Storage||Standard 2000 Pcs, Sample 20000 Pcs|
|Operating Environment||0~40℃, 0~85%RH (no condensing), Altitude < 2000m|
|Storage Environment||-20~50℃, 0~85%RH (no condensing)|
|Standard Accessory||Power Adapter, User Guide, CD Disk(PC Software), USB cable, White and Black Calibration Cavity, Sample holder, 25.4 caliber, 15 caliber, 8 caliber, 4 caliber.|
|Optional Accessory||Micro Printer, Transmission test fixture component, Inspector Green Board|
|Notes: The specifications are subject to change without notice.|
Did you know that your ability to accurately see color depends on many factors? Here’s why your results might be different each time you take the test.
Lighting - Light plays one of the most significant roles in color perception – did you know it's actually the color of the light that determines the color your brain will perceive? You should always judge and compare color samples under controlled lighting conditions, like those provided by a light booth.
Backgrounds Effects - The human eye’s ability to correctly perceive the color is affected by the surrounding colors, a phenomenon known as simultaneous contrast. Always judge color in a neutrally colored environment.
Retinal Fatigue - Our eyes get tired very easily. When we stare at an object for longer than a few seconds, chemicals in our eyes start to deplete and begin sending incorrect information to our brains. When visually evaluating color, you should always rest your eyes between samples.
Poor Color Memory - Our brains find it extremely difficult to remember the specifics of any given color. To effectively determine slight color differences, color samples must be touching.
Age - As we get older, color perception starts to fade. However, color perception isn’t just innate; it's also a learned skill. If you’re responsible for visual color evaluation, you must know your limitations and train your eyes.