Shenzhen ThreeNH Technology Co., Ltd.
Leader Of Color Management In China
|Place of Origin:||China|
|Minimum Order Quantity:||1pcs|
|Packaging Details:||Packed in hard nylon box firstly, and then reinforced with carton box for outer packing|
|Delivery Time:||1 working day|
|Payment Terms:||T/T, Western Union,Paypal|
|Supply Ability:||5000pcs per month|
|Color:||Blue||Measuring Angle:||20° 60°85°|
|Measuring Area:||20°:9X10 60°:9X15 85°:5X38||Measuring Time:||0.5S|
|Accurate:||Conform With JJG696 First Grade Requirements Of Gloss Meter||Long-time Calibration:||Automatically Check And Made The Calibration|
3nh Professional Gloss Meters YG268 0~2000GU used in Bamboo flooring Stainless steel gloss test
Brief Introduction of Precision gloss meter YG268
Professional Gloss Meters YG268 is independently developed by Shenzhen 3nh Technology Co., Ltd with independent intellectual property, manufactured according to international standard ISO2813 and China standard GB/T 9754. With auto-calibration and high-end QC software, it can meet the first grade requirements of JJG696.
YG series gloss meter also features with USB/RS-232 interface and Blutooth 2.1 dual mode.
1. Elegant Design combined with aesthetics and ergonomics;
2. With auto-calibration function;
3. Meet the first grade requirements of JJG696;
4. Large storage to save over 35000 data;
5. Auto Power-off within optional 30-120 seconds;
6. Comply to ISO2813,ASTM523, GB/T9754;
7. With PC software for quality report, more extend functions;
8. With multi working modes and multi-functions for most customers’ requirements.
Widely used in the filed of paints,coating,plastics,ink,rubber,printing,paper, glass, handware, ceramic, marble etc.
More pictures for Precision gloss meter YG268
Specifications for Precision gloss meter YG268
|Model||YG268 Tri-angle Gloss Meter|
|Measuring Angle||20° 60°85°|
|Measuring Area||20°:9X10 60°:9X15 85°:5X38|
|Measuring Range||20°: 0~2000GU|
|Measuring Modes||Basic mode; Statistic mode; Continuous mode; QC mode|
|Accurate||Conform with JJG696 first grade requirements of gloss meter|
|Auto Power-off Time||Within 30s-120s|
|Long-time Calibration||Automatically check and made the calibration|
|Language||Chinese & English|
|Storage||35000(Basic mode &Statistic mode : 1500; Continuous mode : 10000; QC mode: 10000)|
|Display||2.3 inch white and black screen|
|Power Supply||1pc dry-cell battery(can measure 10000 times) or use USB Charge|
|Interface||USB & Bluetooth 2.1 version|
|PC Software||GQC6 QC software can printing testing report and with more extended functions|
|Operation Temperature Range||0-40℃(32-104° F)|
|Storage Temperature Range||(-20-50℃)(-4-122° F)|
|Humidity Range||< 85% RH, without condensation|
|Standard Accessories||USB cable, user manual, calibration plate, QC software|
|Optional Accessories||Miniature printer|
Notification: the technical parameter is only for reference
How to choose gloss meter angle ?
How to choose gloss meter angle ? For early contact with the gloss meter reader is a more confused proposition, it comes out a lot when searching on the Internet. There is single-angle gloss meter, three-angle gloss meter, a 45-degree gloss meter, a 60-degree gloss meter, so which gloss meter is suitable for us to use it?
In general, gloss meter has 20 °, 45 °, 60 °, 75 °and 85 °specifications. Then how to choose gloss meter angles ? If our customers required us what angle gloss meter to measure the gloss data then we need choose according to the customer's request.
If the customer didn’t specify what angle, it can choose 60 °general version gloss meter, 20 °angle meter is generally used in the gloss products higher than 70Gu, 85 °angle is generally used in gloss products less than 10 °, 45 °And 75 °angles used less, 45 °gloss meter is mainly used for the surface gloss measurement of plastic film, 75 °gloss meter is mainly used for enamel surface gloss measurement.
|Gloss Range||60° Value||Notes|
|High Gloss||>70 GU||If measurement exceeds 70 GU, change test setup to 20°|
|Medium Gloss||10 - 70 GU|
|Low Gloss||<10 GU||If measurement is less than 10 GU, change test setup to 85°|
Contact name: Ms. Suzie Xia
Tel: 0086-755-27198826 Ext: 601