Shenzhen ThreeNH Technology Co., Ltd.
Leader Of Color Management In China
|Place of Origin:||P.R.China|
|Certification:||CE / ROHS / FCC / ISO9001:2015|
|Minimum Order Quantity:||1pcs|
|Packaging Details:||Packed in hard nylon box firstly, and then reinforced with carton box for outer packing|
|Delivery Time:||1 working day|
|Payment Terms:||T/T, Paypal, Western Union|
|Supply Ability:||5000pcs per month|
|Division Value:||0.1GU||Measuring Area (mm):||20°: 10x10 60°: 9x15 85°: 5x36|
|Measuring Range:||0~2000GU||Measuring Angle:||20° 60° 85°|
|Measurement:||Basic, Statistical, Continuous||Dimension:||L*W*H : 160mm*75mm*90mm|
3NH Multi Angle Portable Gloss Meter 20 ° 60 ° 85 ° , Tile Floor Gloss Meter Metal 0~2000GU NHG268
NHG268 portable gloss meter is widely used in paint, ink, stoving varnish, coating, wood products; marble, granite, vitrified polished tile, pottery brick and porcelain; plastic, paper; hardware industries, etc. for gloss measurement and gloss data transmission.
|NHG268 Multi-angle Portale Gloss Meter|
|Comply with standard ISO 2813, GB/T 9754, ASTM D 523, ASTM D 2457|
|Measuring Area (mm)||20°: 10x10 60°: 9x15 85°: 5x36|
|Measuring Range||20°:0~2000GU 60°:0~1000GU 85°:0~160GU|
|Measurement Range Repeatability Reproducibility||0-10GU 10-100GU 100-2000GU|
|± 0.1GU ± 0.2GU ± 0.2%GU|
|± 0.2GU ± 0.5GU ± 0.5%GU|
|Measurement Standard||Conform with JJG696 first class glossmeter working requirement|
|Chromaticity Corresponding||Corresponding with CIE 1931(2°) under CIE C light source|
|Errors||±1.2 , ± 1.2%|
|Dimension||L*W*H : 160mm*75mm*90mm|
|Battery||3200mAh Li-ion Battery , >10000 times(within 8 hours)|
|Display||TFT 3.5 inch, capacitive touch screen|
|Storage||Basic mode: 1000; Statistical Mode: 5000, Continuous mode : 5000|
|Software||GQC6 Quality Control Software with QC report printing functions and more extended functions.|
|Humidity||<85% relative humidity, no condensation|
|Standard Accessories||Power Adapter, USB cable, User Manual, CD (Including QC software), Calibration Plate|
|Optional Accessories||Miniature Printer|
|Note||The specifications are subject to change without notice.|
Product detail shows
We aim to provide best services and products in exchange your maximum satisfaction. If you have any dissatisfaction on your purchase(s), please be advised to contact us in advance. We will do our best to resolve your issue till your satisfaction.
Our experienced customer officers will respond your message within 24 hours (exclude weekend and public holidays)
How to choose gloss meter angle ?
How to choose gloss meter angle ? For early contact with the gloss meter reader is a more confused proposition, it comes out a lot when searching on the Internet. There is single-angle gloss meter, three-angle gloss meter, a 45-degree gloss meter, a 60-degree gloss meter, so which gloss meter is suitable for us to use it?
In general, gloss meter has 20 °, 45 °, 60 °, 75 °and 85 °specifications. Then how to choose gloss meter angles ? If our customers required us what angle gloss meter to measure the gloss data then we need choose according to the customer's request.
If the customer didn’t specify what angle, it can choose 60 °general version gloss meter, 20 °angle meter is generally used in the gloss products higher than 70Gu, 85 °angle is generally used in gloss products less than 10 °, 45 °And 75 °angles used less, 45 °gloss meter is mainly used for the surface gloss measurement of plastic film, 75 °gloss meter is mainly used for enamel surface gloss measurement.
|Gloss Range||60° Value||Notes|
|High Gloss||>70 GU||If measurement exceeds 70 GU, change test setup to 20°|
|Medium Gloss||10 - 70 GU||If measurement during 10- 70 GU, change test setup to 60°|
|Low Gloss||<10 GU||If measurement is less than 10 GU, change test setup to 85°|
Shenzhen ThreeNH Technology Co., Ltd
Add: 4/F, Building 8, Nangang Second Industry Zone, Xili, Nanshan District, Shenzhen, 518055 China.
Tel: 0086-755-27198826 Ext: 601